Testers must be familiar and competent in the operation of any and all testing equipment used to carry out sound insulation tests in accordance to industry standards.
|Make||Model||Upload Type||UK Distributor|
|Rion||NA-28||.xlsx||ANV Measurements Systems
|Bruel & Kjaer||2250||.rnd||Bruel & Kjaer
Below is the what standards equipment should be calibrated to. All equipment should be undertaken by a UKAS or iLAC accredited laboratory for the specific class of calibration required.
Traceable, manufacturers or verified calibration is not acceptable.
|Sound Level Meter||Verified against the accuracy class 0 or 1 in IEC 60651 and IEC 60804|
|Sound Level Meter Filters||IEC61260|
|Sound Calibrator||Against accuracy class 1 defined in IEC 60942|
|Reverberation Time||ISO 354|
|Tapping Machine||Annex A of ISO 140-7|
If equipment is hired, a record should be kept of all equipment used for testing and pre-completion testing including that which is not owned by the test.